The programmable constant temperature and humidity aging box (programmable) is used to test various products and materials for heat resistance, cold resistance, high temperature resistance and moisture resistance. Suitable for electronics, photovoltaic modules, electrical appliances, food, vehicles, metals, chemicals, building materials and other factories, research institutes, institutions.
1. Programmable constant temperature and humidity test chamber In the high temperature test, if the temperature change does not reach the test temperature value, the constant temperature and humidity chamber can check the electrical system and eliminate the fault one by one. If the temperature rises very slowly, it is necessary to check the air circulation system to see if the regulating baffle of the wind cycle is normal. On the contrary, check whether the motor of the wind cycle is running normally. If the temperature is too high, you need to set the PID setting parameters. If the temperature rises directly and the temperature is over-protected, then the controller fails and the constant temperature and humidity chamber must be replaced with the control instrument.
2. If the low temperature does not reach the test index, then you should observe the change of temperature, whether the temperature drops very slowly, or the temperature rises after the temperature reaches a certain value. The former should check whether it will be before the low temperature test. Dry the studio, keep the studio dry, then put the test sample into the work room and then test it. If the test sample in the work room is too much, the wind in the workroom cannot be fully circulated. After eliminating the above reasons, It is necessary to consider whether it is a fault in the refrigeration system, so it is necessary to ask the professional of the manufacturer for maintenance. The latter phenomenon is caused by the poor use of the equipment. The ambient temperature at which the equipment is placed and the location where it is placed (the distance between the back of the cabinet and the wall) must meet the requirements (as specified in the operating instructions for the equipment).
3. In the damp heat test, the actual humidity will reach 100% or the actual humidity will be very different from the target humidity. The value is much lower. The former phenomenon may be caused by the drying of the gauze on the wet bulb sensor. Whether the water in the water tank of the ball sensor is lacking, the water level in the water tank is automatically controlled by a water level controller. Check whether the water supply system of the water level controller is normal and the water level controller works normally. It may be that the wet ball gauze is hardened due to the long use time or the purity of the water supply, so that the gauze becomes hard and the gauze cannot absorb moisture and is dry. The above phenomenon can be eliminated by replacing or cleaning the gauze. The latter phenomenon is mainly because the humidification system does not work. Check the water supply system of the humidification system, whether there is a certain amount of water in the water supply system, whether the water level control of the humidification boiler water level is normal, and whether the water level in the humidification boiler is normal. If everything above is normal, then check the electrical control system, which should be repaired by professional maintenance personnel.
4. When the programmable constant temperature and humidity test chamber suddenly fails during the test operation, the corresponding fault display prompt appears on the control meter and there is an audible alarm prompt. The operator can quickly check which type of fault belongs to the fault diagnosis chapter in the operation of the equipment, and then ask the professional to quickly eliminate the fault to ensure the normal operation of the test. Other environmental testing equipment will have other phenomena in use, and it will be specific phenomena, specific analysis and elimination.
The programmable constant temperature and humidity test chamber should also be regularly maintained, the condenser of the refrigeration system should be cleaned regularly, the moving parts should be lubricated according to the instructions, the electrical control system should be regularly maintained and checked, etc. These tasks are indispensable. The reason why the programmable constant temperature and humidity test chamber is easy to catch a cold
ASLI research and development, production, sales, maintenance: programmable constant temperature and humidity test chamber, thermal shock test machine, programmable constant temperature and humidity test machine, high and low temperature box, three boxes (two boxes) type of hot and cold Impact testing machine, temperature and humidity rapid temperature change box, high temperature test box, salt spray corrosion box, walk-in constant temperature and humidity chamber, ultraviolet weathering test box, xenon lamp weathering test box, vibration, drop, tensile testing machine, battery testing equipment, etc. Simulated test equipment. Products are widely used in electronics, optoelectronics, automobile and motorcycle industry, electronic system factory, electronic parts, automotive electronics, painting industry. Biochemical technology, pharmaceutical industry, state-owned enterprises, military industry and other industries
Customized Wardrobe,Tempered Glass Wardrobe,Bedroom Wooden Wardrobe,Large Closet Wardrobe
Foshan Artcheer Home Solution Corporation Limited. , https://www.artcheer88.com